Fluke 8846A - Closed Loop + Adjustments
ProCal procedure for Fluke 8846A, includes automatic adjustment points.
This procedure automatically calculates uncertainties for each test point.
UUT control over GPIB only
Test points to manufacturers specifications with the following exceptions:
100V Range - 100kHz point performed at 20V
750V Range - 20kHz point performed at 200V
100V and 750V High-Frequency calibration is not performed, however, full-scale gain is performed
- National Instruments GPIB Interface
- Transmille 3041A or Better (Better TUR Ratios will be achieved with 3010A or 4010)
- Suitable Connection Leads
- ProCal Version 5.19 or later.